When installing the Pressure Gauge, it is necessary to […]
When installing the Pressure Gauge, it is necessary to conduct a comprehensive analysis of the installation environment, so as to carry out a reasonable installation of the Pressure Gauge according to the actual installation environment to ensure that the Pressure Gauge is in a stable and safe environment. According to the installation specifications of the Pressure Gauge, you need to pay attention to the following aspects during installation:
Pressure Gauge needs to be installed in an easy-to-observe position: When choosing the installation position of Pressure Gauge, it is necessary to choose an easy-to-observe position for installation so that the Pressure Gauge can be fully observed and dirt treated. In order to improve the accuracy of the Pressure Gauge, it is necessary to install the Pressure Gauge in a location that is not exposed to direct sunlight and high temperature to avoid radiant heat, freezing and vibration of the Pressure Gauge due to external factors.
The Pressure Gauge body needs to be in the vertical direction: When the Pressure Gauge body is installed, the Pressure Gauge body needs to be controlled in the vertical direction so that the staff can read the displayed value of the Pressure Gauge. If the Pressure Gauge needs to be installed at a higher position, it can be adjusted vertically by tilting it forward, but the tilt angle cannot exceed 30 degrees.
Install the buffer tube in front of the Pressure Gauge: After the installation of the Pressure Gauge is completed, a buffer tube needs to be installed in front of the Pressure Gauge to prevent the Pressure Gauge from being affected by external air. As a safeguarded facility of Pressure Gauge, the buffer tube can relieve the gas in the spring tube, and the buffer tube has a certain condensate collection function, which can prevent the Pressure Gauge from excessive gas.
Install the three-way cock or type valve: At the bending part of the Pressure Gauge buffer tube, it is necessary to install the three-way cock or type valve at a suitable intermediate position, so that the three-way cock or type valve can ensure the safety of the Pressure Gauge work.
Cixi Oukate Instrument Technology Co., Ltd. is a pressure gauge manufacturer and gas test gauge and other products. Welcome to visit our official website.